Please choose a body region on the right for you to pin point the problem area of your body.
Shop by Condition
Shop by Brand
Item # ADT-413DT | SKU # ADC-413B
Get an accurate temperature read with our popular compact digital stick thermometer, the Adtemp II 413 Thermometer by ADC that weighs only 11 grams. An audible tone is made when measurement if complete.
Features for this stick thermometer include:
Model No. 413B
Specifications for the
|Range °F:||90.0 to 109.9|
|Range °C:||32.0 to 42.9|
|Response Time Oral:||30 - 40 sec|
|Response Time Rectal:||25 - 35 sec|
|Response Time Axillary:||90 sec|
|Battery Type:||1.55V (LR41)|
|Battery Life:||4,500 uses|
|Dimensions:||5" x 3/4" x 3/8" (12.7 cm x 1.905 cm x .9525 cm)|
|Weight:||0.388 oz (11 g)|
|Operating Environment:||50°F to 104°F (10°C to 40°C)|
|Storage Environment:||-13°F to 131°F (-25°C to 55°C)|
How to Use:
Switching Temperature Modes
From the OFF position, the Celsius or Fahrenheit mode can be switched by pressing and holding the multifunction button for at least 5 seconds. After 3 seconds in one scale the °F or °C will flash, indicating ready to measure.
Ensuring Accurate Temperature Measurement
We recommend using probe covers with all digital stick thermometers (Figure 1).
Oral use: Place the probe well under the patient’s tongue (Figure 2). Keep mouth closed while thermometer is reading. Instruct children not to bite down on the probe. The peak temperature should be reached in as little as 30* seconds. Opening the mouth or improper placement of the probe tip could result in a longer time for a reading. (*30 second response time achieved in water bath test)
Rectal Use: Lubricate the probe cover with a water-soluable jelly for easier insertion; do not use petroleum jelly, as it may cause an inaccurate reading. Gently insert the tip of the probe no more than 1/2 inch into the rectum.
STOP if you meet any resistance. The sensing unit is on the very tip of the probe and there is no need to insert the probe deep into the rectum. Dispose of the used probe cover and wash the thermometer as recommended. The peak temperature should be reached in approximately 25-35 seconds.
Axillary Use: Wipe armpit with a dry towel. Place the probe in the patient’s armpit (Figure 3) and keep the patient’s arm pressed firmly against the body. In order to achieve a better axillary reading, a minimum measuring time of 90 seconds or longer is recommended, depending on placement of the probe and closure of the arm against the body.
Cleaning and Disinfecting
Do not use corrosive detergent to clean the unit.
Clean and disinfect the thermometer before and after each use.
1. Wipe probe with a cloth dipped in a disinfectant solution such as 70% isopropyl (rubbing) alcohol or soap and water.
2. Rinse off disinfectant residue.
Care and Storage